MadgeTech continues to evolve its product line with one mission in mind: to simplify data logging solutions for easy validation. With the newly redesigned TCTempXLCD, view readings in real-time and directly download the data to the MadgeTech 4 Software for reporting, no interface needed!Continue reading Multitasking Done Right with the Newly Resigned TCTempXLCD
MadgeTech Blog — thermocouple
RTD and thermocouple probes are both used to measure temperature, but which one is right for your application? To help decide which will perform best based on specific needs, there are three main questions to ask.Continue reading RTD vs. Thermocouple
Can I use a Thermocouple Simulator to calibrate my thermocouple based data logger?
Thermocouple calibrators do not work reliably with most MadgeTech thermocouple loggers. The data logger will pull down the voltage of the thermocouple calibrator enough to alter the reading. While the thermocouple calibrator will recover quickly, the data logger will take the reading before the calibrator has corrected itself. This will lead to an inaccurate reading. Due to this, using a thermocouple simulator is not recommended when calibrating MadgeTech loggers.